- Original article
- Open Access
3D convolutional neural networks applied to CT angiography in the detection of acute ischemic stroke
European Radiology Experimental volume 3, Article number: 8 (2019)
The aim of this study was to investigate the feasibility of ischemic stroke detection from computed tomography angiography source images (CTA-SI) using three-dimensional convolutional neural networks.
CTA-SI of 60 patients with a suspected acute ischemic stroke of the middle cerebral artery were randomly selected for this study; 30 patients were used in the neural network training, and the subsequent testing was performed using the remaining 30 patients. The training and testing were based on manually segmented lesions. Cerebral hemispheric comparison CTA and non-contrast computed tomography (NCCT) were studied as additional input features.
All ischemic lesions in the testing data were correctly lateralized, and a high correspondence to manual segmentations was achieved. Patients with a diagnosed stroke had clinically relevant regions labeled infarcted with a 0.93 sensitivity and 0.82 specificity. The highest achieved voxel-wise area under receiver operating characteristic curve was 0.93, and the highest Dice similarity coefficient was 0.61. When cerebral hemispheric comparison was used as an input feature, the algorithm performance improved. Only a slight effect was seen when NCCT was included.
The results support the hypothesis that an acute ischemic stroke lesion can be detected with 3D convolutional neural network-based software from CTA-SI. Utilizing information from the contralateral hemisphere appears to be beneficial for reducing false positive findings.
This is the first study applying three-dimensional convolutional neural networks (3D CNN) to computed tomography angiography (CTA) source images for ischemic stroke detection
Stroke detection was improved when cerebral hemispheric comparison and non-contrast computed tomography (NCCT) were included in the CNN analysis.
High sensitivity and specificity in the detection of stroke lesions was achieved
An acute ischemic stroke is caused by thrombotic or embolic occlusion of a cerebral artery. Occlusion of the proximal cerebral artery causes a deep ischemia resulting in a collapse of cellular energetics. This is followed by a necrotic cell death in few minutes. The infarct core is surrounded by a partial area of ischemia, penumbra, where neurons will die within hours. Accurate identification of this “tissue at risk” could be used to identify patients who would benefit most from treatment . In a large ischemic stroke, the extent of damage will increase during the following days. In an extreme case, the mass effect coupled with the tissue damage leads to an increased intracranial pressure and a loss of cerebral blood circulation.
The restoration of blood flow to the tissue can be achieved by opening the embolic artery. An essential aim of acute treatment of brain infarction is to restrict the size of the infarct by a rapid and permanent recanalization of the obstructed artery. Primary imaging methods for the acute ischemic stroke are non-contrast computed tomography (NCCT), CT perfusion (CTP), and CT angiography (CTA). The advantage of CT-based methods is that they are rapid and more widely available than MRI in the emergency setting.
The NCCT imaging must be performed as soon as possible after the stroke code has been activated . NCCT is highly sensitive for the depiction of hemorrhagic lesions  or other NCCT visible mimics of the stroke. The crucial role of NCCT is also the detection of ischemic signs of established ischemic lesion. The main NCCT finding is a cortical-subcortical hypoattenuating area within a vascular territory. It is, however, well-known that NCCT has a relatively low sensitivity in the first 24 h, especially within the limited (3–6 h) time window for thrombolytic treatment .
The extension of the acute infarct is typically estimated from CT perfusion parameters (mean transit time, cerebral blood volume, and cerebral blood flow). In CTP, alterations in cerebral blood flow and volume can be seen in the acute stroke patients. However, many multi-detector scanners still offer limited coverage for CTP of the brain which is often stated as a drawback of CTP. Newer 256-slice scanners can provide whole-brain coverage, and in the next few years this coverage will most likely be widely available . CTP implies a relatively high radiation exposure and coarser resolution and thicker slices when compared, for example, to CTA.
Using CTA, it is possible to demonstrate the anatomy of the aortic arch, carotid, and cerebral arteries, the presence of stenosis or thrombus of carotid or cerebral arteries. In addition, an impression of the functioning collateral network is obtained. Hypoattenuation in CTA source images (CTA-SI) showing a lack of enhancement provides an estimate of the cerebral blood flow reduction, whereas the NCCT allows to assess changes in brain tissue water content. In comparison, CTP is performed by monitoring only the first pass of an iodinated contrast agent bolus through the cerebral circulation . Interestingly, CTA-SI can provide another way, complementary to CTP, to evaluate the size of acute infarct [6, 7], even though this method is not typically used in clinical practice. CTA detects the regions of the brain with hypoattenuation due to long delays in contrast arrival to the ischemic brain tissue. It has been discussed in several studies (e.g., ) about CTA correlating more with a CTP cerebral blood flow map (penumbra and core) than with the corresponding cerebral blood volume map .
Among the CT techniques, CTA has multiple advantages. It is highly specific and more sensitive than NCCT in the detection of early irreversible ischemia . In addition, it provides the possibility to evaluate the whole brain vasculature which is not available with CT perfusion. The clinical outcome following the recanalization has been discussed in other studies, and it is believed to be strongly dependent on the extent of the pretreatment infarct lesion volume . The total stroke lesion volume can be used as a predictor of treatment efficacy. A pre-treatment infarct volume lower than 70 cm3 can be set as threshold for predicting a good clinical response to reperfusion .
Applying artificial intelligence to stroke sign detection can potentially aid in diagnosis or predicting prognosis . One of the motivations in applying machine learning and deep learning methods to medical images is the automatic extraction of non-trivial and non-linear features from the imaging data. These features can then be applied for example in tissue classification or lesion volumetry. If the imaging data are inherently three-dimensional (3D), it is natural to use methods based on 3D features. One such AI algorithm family is 3D convolutional neural networks (CNNs). They increase computational costs and hardware requirements compared to their two-dimensional counterparts. However, parallel processing, especially utilization of graphics processing units, as well as the recent advancement in deep learning algorithms, have allowed 3D CNN-based software tools to take further steps toward clinical deployment.
CNNs and their application in medical imaging have been discussed in length by other researchers [13,14,15]. A review of AI application areas in stroke imaging have been written by Lee et al. . Recently, Lisowska et al.  studied the detection of dense vessels and of ischemia from NCCT images using CNNs.
An automatic tool to detect infarcted regions from brain CTA-SI could aid in diagnosis and evaluate the infarct extent or volume.
This study had two aims: (1) to investigate if using a 3D CNN is feasible for detecting and segmenting hypoattenuated regions in CTA-SI of stroke patients; and (2) to investigate if analysis performance can be improved by including NCCT and information from contralateral hemisphere as inputs to the neural network. The training and evaluation of the networks were based on manual segmentations of the hypoattenuated regions. This is the first study investigating the application of 3D CNN on CTA-SI for acute ischemic stroke lesion detection.
Subjects and imaging parameters
The clinical and imaging findings of the consecutive stroke suspected cases of the middle cerebral artery or/and internal carotid artery occlusion presented to Helsinki University Hospital between November 2016 and December 2016 were retrospectively reviewed, and 60 randomly selected patients were included in the study. All of them had undergone an identical CT protocol. Thirty patients had been diagnosed by neurologists with an acute ischemic stroke (group A). The CTA and CTP of the other 30 patients showed no signs of a stroke (group B). Both groups were randomly selected from the corresponding patient populations (the group A was limited to the patients with unilateral ischemic strokes of the middle cerebral artery). Inclusion criteria were (1) stroke code activated; (2) admission stroke protocol imaging performed using fast CTA-SI acquisition protocol; and (3) documented middle cerebral artery or/and internal carotid artery occlusion on CTA-SI.
In the group A, the median (range) National Health Stroke Scale (NIHSS) was 17 (3–27). Disease burden in the group A consisted of total of 15 patients which were earlier diagnosed with a cardiovascular disease (atrial fibrillation and/or coronary artery disease), four had a type 2 diabetes, four had a diagnosis of breast cancer, and two had a diagnosis of chronic obstructive pulmonary diagnosis. The rest of the group A were disease-free before the acute stroke. In the group B, the patients presented with symptoms suggestive of stroke (focal neurological symptoms with sudden onset). The group B was randomly selected from patients imaged with the stroke protocol who did not have a diagnosis of stroke. Seventeen patients in the group B underwent a follow-up brain MRI in the following 48 h from stroke code activation without stroke diagnosis. The rest of the stroke-negative group B were domiciliated by the neurologists during the next 24 h without the follow-up scan and without diagnosis of stroke.
In the group A, the number of patients with occluded middle cerebral artery segments (M1–4) and internal carotid artery (ICA) were M1 (n = 16); M2 (n = 5); M3 (n = 1); M4 (n = 1); ICA (n = 4); ICA + M1 (n = 1); and ICA + M2: (n = 2). The symptoms had begun ≤ 1 h (n = 8), 1–2 h (n = 7), 2-3 h (n = 2), and ≥ 3 h (n = 12) prior to the CT in the group A. The group A consisted of 15 males and 15 females while the group B consisted of 14 male and 16 female patients. The median (min-max) ages were 73 (47–82) years and 63 (26–91) years in the groups A and B, respectively. All the patients were imaged using a Somatom Definition Edge (Siemens Healthineers, Erlangen, Germany) 128-slice CT scanner, with the same protocol described in Table 1. CTA scanning range included the carotid arteries. NCCT and CTP were limited to the intracranial range. Due to the retrospective nature of the study, only transversal reconstructions with slice thicknesses/increments of 0.75/0.5 mm (CTA) and 4/4 mm (NCCT) were available for the analysis described below. Due to the manually adjusted field-of-views, nominal in-plane resolutions were from 0.32 × 0.32 mm2 to 0.55 × 0.55 mm2 (fixed 512 × 512 matrix). All data were anonymized and stored on a server running the Extensible Neuroimaging Archive Toolkit, or XNAT, version 1.1.6 .
A senior neuroradiologist and a radiologist, with over 20 and over 5 years of experience, respectively, segmented the infarct regions on the CTA-SI for the 30 stroke-positive patients (group A), in consensus, only including the visible hypoattenuated regions. No manual delineation was done for the group B. The overview of image processing workflow is presented in Fig. 1.
The CNN implementation had three requirements for the input data. Firstly, patient’s CT and CTA image series were cropped to equal image volume dimensions. Secondly, all the data over all the patients needed to have the same image resolution. Thirdly, the data was recommended to be zero mean, and unit variance normalized to aid the CNN to converge.
The CTA volumes (which included carotid arteries and neck region) were cropped to include only the brain region. This was done to decrease memory requirements. To include information from the contralateral hemisphere, an additional set of images was created by flipping the left and right sides of the original CTA-SI, hereafter called hemispheric comparison volume. This allowed for the inclusion of the information from the approximate contralateral anatomical regions for the CNN training and evaluation. The flipped images and the NCCT images were then matched to the original images using rigid registration. The image co-registrations were done using BRAINSFit tool . Approximate intracranial spaces were identified using an in-house MATLAB (MathWorks, Inc., Natick, MA, USA) script. Then, the data were scaled to zero mean and unit variance in the brain regions. Finally, all the volumes across the data set were resampled to 0.5 × 0.5 × 0.5 mm isotropic resolution. Convert 3D (part of ITK-SNAP toolkit ) command-line tool was used for rescaling and resampling. The consistent voxel size across the data and the intensity shift and rescaling were required for the CNN framework in use. Within each subject, the volumes, including the manual lesion delineations via the nearest neighbor resampling, were in the same image space. All the pre-processing tools had command-line interfaces allowing easy automation and preprocessing multiple subjects in parallel.
All images were eventually verified visually to confirm the successful registrations. 3D Slicer image processing and visualization platform  was used in the manual segmentation and the visual verification of the preprocessing outcomes.
Convolutional neural network
DeepMedic , an efficient 3D CNN available as a free open-source implementation based on Theano library , was used in the machine learning of the infarct features. The multi-scale network was 11-layers deep with a batch size of 10, L1 regularization 10−6 and L2 regularization 10−4. Root mean square propagation optimizer (ρ = 0.9, ε = 10− 4) was used with Nesterov momentum value 0.6. Training was done over 35 epochs with 15 sub-epochs each. Initial learning rate was set to 0.001 and halved at eight predefined epochs. To remove left-right directional dependence from the trained model, during CNN training, the data were augmented by reflecting all the images along sagittal axis. This step mirrors all the images in unison and is therefore not related to the hemispheric difference volume mentioned above.
The patient groups (A and B) were partitioned randomly into two sets: training set (15 from A + 15 from B) and testing set (15 + 15). The testing set was separate from the training phase and was used only in the reporting of the results. An intracranial mask was used to limit the investigative volume. The manually drawn stroke volumes were considered as ground truths. The output consisted of a single label, i.e., the estimated infarct lesion.
Without changing the data set partitioning or network parameters, three different training runs were performed: (1) CTA as a single input; (2) CTA and left-right flipped CTA for cerebral hemispheric comparison as two inputs; and (3) CTA, flipped CTA for cerebral hemispheric comparison, and NCCT as three inputs. All three CNN outputs were evaluated with the same testing data. The CNNs produced confidence estimates (or probability maps) valued from zero to one for the infarcted regions. In addition to voxel-wise segmentation, clinically relevant anatomical regions were recorded as stroke-positive or stroke-negative. Standard anatomical regions used by the Alberta stroke program early CT scoring system (ASPECTS) were chosen as these regions. If a CNN produced even a single positive voxel in the ASPECTS anatomical region, the region was defined as stroke-positive.
All preprocessing, training, and testing were done on a Linux (Debian 8.5) workstation with Intel Xeon E5-2687 W v4 3.00 GHz processor, 64GB RAM and NVIDIA Quadro M6000 24GB graphics processing unit.
The CNN performance was evaluated against the expert segmentation by two approaches, i.e., by anatomical regions visually and in a volumetric voxel-wise manner. The predefined anatomical regions according to the ASPECTS  were six segments of middle cerebral artery (M1–M6); insular ribbon (I); nucleus lentiform (L); caudate (C); and internal capsule (IC). These were visually marked as stroke-positive or stroke-negative in the ground truth data and in the three CNN outputs. This approach was chosen to emphasize clinically relevant brain subregions in the evaluation. Secondly, full-brain voxel-wise overlaps between the outcomes and ground truth were compared by calculating Dice similarity coefficients (DSC) and infarct volumes. On all CNN outputs, a conservative threshold probability value of 0.5 was used for labeling voxels as positive. Receiver operating characteristic (ROC) curves were calculated for the voxel-wise overlap by varying the threshold value, and the areas under the curves (AUC) were calculated. The extent and location of the false positive outputs of the CNNs were re-investigated visually by both radiologists. This was done to estimate the possible reasons for false positives. ASPECTS scorings for manual and CNN segmentations of the CTA-SI were calculated for the 30 group A test patients, to concretize the individual differences between the three CNN options.
CNN training is a computationally demanding process, whereas creating ground truths by expert segmentation is often expensive, tedious, and time consuming. To give an oversight of these expenses, CNN training durations were recorded, and one subject was chosen from the group A for which the manual segmentation, automatic preprocessing run, and CNN execution times were recorded.
Representative CTA-SI slices with manual and one of the CNN segmentations are shown in Fig. 2 for nine patients of group A. The CNN false positives are shown in Fig. 3 for nine patients of group B. Ten individual anatomical regions (according to the ASPECTS, left and right hemisphere) were manually labeled as positive or negative in the three CNN outputs (Table 2). The manual segmentations were considered as the ground truths. The Dice similarity coefficient (DSC) was calculated from the regions’ true or false labeling (see Table 2).
As explained in the “Methods” section, three separate CNNs were trained: (1) with CTA-only; (2) with CTA + cerebral hemispheric comparison; and (3) with CTA + cerebral hemispheric comparison + NCCT as inputs. Using a 0.5 threshold on probability maps voxel-wise, sensitivity was 0.67, 0.74, and 0.71; specificity 0.93, 0.96, and 0.96; and DSC 0.40, 0.55, and 0.55 for the three CNNs, respectively. The threshold was varied, showing the resulting ROC curves in Fig. 4. The ROC-AUC was 0.91, 0.93, and 0.93, respectively. These included all the voxels from the group A and B testing data. Variability between stroke patients was estimated from the group A testing data. The respective means ± standard deviation of the patient-wise ROC AUC was 0.89 ± 0.04, 0.92 ± 0.06, and 0.91 ± 0.05. A small improvement (e.g., combined voxel-wise sensitivity increased by 0.07 and DSC by 0.15) was seen when the cerebral hemispheric comparison CTA was included, but there was practically no effect when the NCCT was added as a third input. The combination of CTA and NCCT was not investigated. The largest voxel-wise DSC value (0.61) was obtained with CTA + cerebral hemispheric comparison CTA inputs at 0.7 threshold.
During the retrospective visual investigation of the false positives, the appearance of unilateral and bilateral false positives in the CNN outputs (Table 2, Figs. 2 and 3) was linked with the age-related periventricular white matter hypoattenuation (see Fig. 3b, c). Cortical false positives were related to the widening of the cortical sulci due to cortical atrophy (see Fig. 3d). As a consequence, most of the false positives were located in the periventricular white matter and near to cortical cerebrospinal fluid spaces, to be associated with the normal aging effects on the brain.
Lesion volume comparisons between the manual delineations and CNN outputs (probability threshold 0.5) were calculated from the largest continuous lesion segments. The infarct lesion volumes in the manual segmentations varied from 11 to 293 cm3. The comparisons are shown in Fig. 5.
Neither manual interventions nor modifications had to be made for any of the volumes when visually verifying the image preprocessing. Training the CNNs using the 30 subjects took the total of 77 h for CTA-only, 122 h for CTA + cerebral hemispheric comparison-CTA, and 152 h for CTA + cerebral hemispheric comparison-CTA + NCCT input features.
The time for processing and CNN execution steps in an individual subject was as follows: 23 min for manual segmentation including data selection and saving; 2 min and 1 s for the file compression to save disk space; 3 min and 26 s for executing the CNN for CTA only; 3 min and 36 s for executing the CNN for CTA + cerebral hemispheric comparison; 3 min and 40 s executing the CNN for CTA + cerebral hemispheric comparison + NCCT features. This time included initialization and reading of the network weights, calculating outputs, and saving the results. Transfer from the picture archive, anonymization, and file type conversions was not included.
The aim of this study was to investigate the feasibility of ischemic stroke detection from CTA-SI using 3D CNNs. All the lesions in the testing data were correctly lateralized, and a high correspondence to manual segmentations was achieved. Although only a minor effect was seen on the ROC-AUC values, cerebral hemispheric comparison CTA as an additional input was beneficial in improving the ischemic lesion detection specificity. A substantial increase in specificity can be seen when cerebral hemispheric comparison CTA was included as an input feature. A slight decrease in the number of false positives was seen when NCCT was included. No notable changes can be seen in sensitivity between the three outputs. A notable change was seen when the cerebral hemispheric comparison CTA was included and a very slight change for the worse with the addition of the NCCT input. A notable improvement to the average performance was seen when the cerebral hemispheric comparison CTA was included as an input feature. However, a general impression of overestimating small lesion volumes and underestimating large volumes compared to the manual segmentation was seen for all trained networks (Fig. 5). This can also be seen in the ASPECTS scores defined from the CTA-SI using manual segmentation and the three CNNs (Fig. 6): at low (manual) APSPECT scores (i.e., large lesions), the CNN scores were overestimated and vice versa for higher manual ASPECTS scores (smaller more localized lesions).
CNNs have previously been applied in the ischemic stroke detection based on diffusion-weighted imaging (DWI)  and multi-spectral MRI . Lisowska et al.  proposed using a CNN in dense vessel and ischemia detection from NCCT images. The ischemia detection ROC-AUC values were in line with the present study (0.92 and 0.93, respectively). Also in line was the usefulness of comparing hemispheres in the detection of ischemia. However, Lisowska et al. demonstrated a more substantial absolute increase in ROC-AUC (+ 0.18) than the present study (+ 0.02). Additionally, they included an anatomical context by atlas registration but for ischemia detection found it less important than the direct bilateral comparison. In order to include contralateral features and anatomical context, they designed a custom network architecture where left and right hemisphere features propagated in parallel layers. The current study demonstrated that hemisphere information can be successfully included by using a co-registered left-right flipped image as an additional data channel. This enabled the use of the 3D CNN framework originally designed for lesion detection from multi-channel MRI .
There are other architectural differences to the DeepMedic used in the present study. Primarily the CNN by Lisowska et al.  used a composition of 1D convolutions, whereas the DeepMedic is based on fully 3D convolutions. Maldjian et al.  and Nowinski et al.  have also utilized a cerebral hemispheric difference in their analysis workflows. However, they did not use CNNs. Sensitivity calculated from Maldjian et al. reported values for lentiform nucleus and insula (0.85) lower than the sensitivity calculated over the ASPECTS regions in the present study (0.93). Gillebert et al.  as well as Peter et al.  applied a texture and atlas-based approach to lesion segmentation. Gillebert et al.  found a slightly higher DSC (0.67) than the present study (0.61 at optimal threshold) for simulated data but reported high variance for performance using actual patient images. Peter et al.  reported a ROC AUC of 0.82 which was less than that of the present study. At least one infarct detection software has been made commercially available .
The above mentioned studies had NCCT images as inputs whereas the current study is the first one to apply a 3D CNN to CTA-SI. In the previous studies, difficulties in distinguishing between old and new stroke lesions, and age-related changes have been reported. Likewise, in this study, the areas with periventricular white matter hypoattenuation and cortical atrophy were mislabeled as stroke lesions. The severity of this problem was mitigated by the introduction of cerebral hemispheric comparison CTA, and to a lesser degree, NCCT in addition to hemispheric comparison CTA, as input features. However, bilateral ischemic strokes were not investigated. Detection of bilateral stroke might not benefit from the proposed cerebral hemispheric comparison method.
Volumetric diagnosis of the ischemic lesion on a CTA-SI versus DWI and CTP lesion volumes is a controversial topic. The studies using single- and 4-slice CT scanners showed a high correlation between CTA-SI and DWI lesion volume in acute stroke [31, 32]. CTA-SI acquired with a fast-acquisition protocol overestimates the volume of the infarct core compared to DWI. According to Yoo et al. , the lesion overestimation can be significant (> 25 cm3). Major disadvantage of the DWI is its limited availability in the emergency setting. On the other hand, the inclusion of CTP increases substantially the radiation exposure and the contrast agent doses. Our stroke protocol consisted of NCCT and CTP imaged before CTA test-bolus and CTA-SI. According to a recent review by Sharma et al. , despite the theoretic modeling suggesting CTA-SI to be predominantly blood-volume weighted, the more rapid modern CTA protocols may be too quick to achieve the arterial and tissue contrast steady-state. In addition, these protocols are more likely blood-flow-weighted.
The hypothesis of this study was also that CTA imaged precisely after the test-bolus the extent of ischemic core on CTA-SI is proportional to the cerebral blood flow (penumbra and core) deficit justifying the proposed deep learning approach. However, as can be seen in Fig. 5, there was a relatively high variance in the measured volumes between the manual and CNN segmentations, which makes the use of any definite volume thresholds problematic. One explaining factor for the discrepancy may be the objective difficulty of delineating the infarct region accurately. It may be partially responsible for the mismatch between the volumes for small lesions (whose volume was overestimated by CNN) and large lesions (whose volume was underestimated by CNN). On the other hand, the varying radiological manifestation of different infarct types may contribute to the neural network performance. This underlines the importance of using large training datasets.
The implementation of this study was primarily based on freely available open-source software and the steps could be combined into an automated stroke analysis pipeline. Low latency processing in a time critical situation where stroke imaging is usually applied is vital. The total analysis taking currently less than 6 min for an individual patient is approaching an acceptable level of delay. No optimization of network model or training parameters, e.g., adjusting the learning rate, was performed for these results. It should also be noted that the time required for network transfers was not included in the estimates. Transfer delays could affect the total time significantly. Including the cerebral hemispheric comparison CTA as an input had only a miniscule effect on the processing time. DeepMedic implementation of a well-established Theano framework was chosen due to its previous success in multi-contrast MRI brain lesion segmentation challenge . The present study showed a satisfactory performance for a different task and for different modality demonstrating the power of the network design in this “off-the-self” implementation.
The typical problem of model overfitting can be mitigated with a sufficiently representative selection of training data. These data should incorporate the variability in target manifestations as extensively as possible. The dataset comprised of different stroke severities, lesion volumes, symptom starting times, and various occluded arteries. This resulted in a small sampling of individual types. Stroke can manifest in multiple ways and therefore the small training set was a study limitation. Future improvements of the analysis workflow and larger data set are warranted for including the variety of the disease manifestations.
In addition, image resolution varied between data sets, which introduced additional variance. The effect was mitigated by using a unified resampling resolution. In a case of wide clinical application, the tool should remain robust against varying imaging parameters, including resolution. A similar limitation was the lack of thin slice NCCT. Although including the NCCT (along with CTA and hemispheric comparison CTA) had only a minor noticeable effect (small decrease in the false positive rate of non-infarcted patients), the authors believe that including both investigated additional feature inputs is a worthwhile pursuit. Moreover, NCCT alone or CTA + NCCT without hemispheric comparison were not investigated. Data processing could be further improved, by including atlas-based anatomical regions as described by Peter et al.  or by introducing additional features such as the requirement of hypoattenuation to extend onto the cortical surface. In addition, finding an optimum in the CNN parameter space, exploring alternative algorithms and implementations aimed at improving specificity and reducing processing time is essential when targeting the clinical deployment. Finally, a fully independent validation data from another imaging site and from a different vendor CT equipment should be used to assess the transferability and robustness of the trained network.
In conclusion, the detection of acute ischemic brain lesions by deep learning analysis of CTA-SI was studied and high sensitivity was achieved. Improvement in specificity was seen when contralateral hemisphere was included in the analysis. These encouraging findings demonstrate a potential of CNNs for a rapid, precise, and fully automated method to assist radiologist and clinicians in the detection of acute stroke lesions.
Alberta stroke program early CT scoring system
Convolutional neural network
Computed tomography angiography
CTA source images
Computed tomography perfusion
Dice similarity coefficient
Non-contrast computed tomography
- ROC AUC:
Area under receiver operating characteristic curve
Murphy BD, Fox AJ, Lee DH et al (2006) Identification of penumbra and infarct in acute ischemic stroke using computed tomography perfusion-derived blood flow and blood volume measurements. Stroke 37:1771–1777
Wardlaw JM, Seymour J, Cairns J, Keir S, Lewis S, Sandercock P (2004) Immediate computed tomography scanning of acute stroke is cost-effective and improves quality of life. Stroke 35:2477–2483
Kucinski T (2005) Unenhanced CT and acute stroke physiology. Neuroimaging Clin N Am 15:397–407
de Lucas EM, Sánchez E, Gutiérrez A et al (2008) CT protocol for acute stroke: tips and tricks for general radiologists. 28:1673–1687
Hoeffner EG, Case I, Jain R et al (2004) Cerebral perfusion CT: technique and clinical applications. Radiology 231:632–644
Shetty SK, Lev MH (2005) CT perfusion in acute stroke. Neuroimaging Clin N Am 15:481–501
Schaefer PW, Yoo AJ, Bell D et al (2008) CT angiography-source image hypoattenuation predicts clinical outcome in posterior circulation strokes treated with intra-arterial therapy. Stroke 39:3107–3109
Sharma M, Fox AJ, Symons S, Jairath A, Aviv RI (2011) CT angiographic source images: flow- or volume-weighted? AJNR Am J Neuroradiol 32:359–364
Camargo EC, Furie KL, Singhal AB et al (2007) Acute brain infarct: detection and delineation with CT angiographic source images versus nonenhanced CT scans. Radiology 244:541–548
Lev MH, Segal AZ, Farkas J et al (2001) Utility of perfusion-weighted CT imaging in acute middle cerebral artery stroke treated with intra-arterial thrombolysis: prediction of final infarct volume and clinical outcome. Stroke 32:2021–2028
Yoo AJ, Hu R, Hakimelahi R et al (2012) CT angiography source images acquired with a fast-acquisition protocol overestimate infarct core on diffusion weighted images in acute ischemic stroke. J Neuroimaging 22:329–335
Lee EJ, Kim YH, Kim N, Kanga DW (2017) Deep into the brain: artificial intelligence in stroke imaging. J Stroke 19:277–285
Suzuki K (2017) Overview of deep learning in medical imaging. Radiol Phys Technol 10:257–273
Shen D, Wu G, Suk HI (2017) Deep learning in medical image analysis. Annu Rev Biomed Eng 19:221–248
Litjens G, Kooi T, Bejnordi BE et al (2017) A survey on deep learning in medical image analysis. Med Image Anal 42:60–88
Lisowska A, O’Neil A, Dilys V et al (2017) Context-aware convolutional neural networks for stroke sign detection in non-contrast CT scans. In: Valdés Hernández M, González-Castro V (eds) Medical Image Understanding and Analysis. MIUA 2017. Communications in Computer and Information Science, vol 723. Springer, Cham
Marcus DS, Olsen TR, Ramaratnam M, Buckner RL (2007) The extensible neuroimaging archive toolkit: an informatics platform for managing, exploring, and sharing neuroimaging data. Neuroinformatics 5:11–34
Johnson HJ, Harris G, Williams K (2007) BRAINSFit: mutual information registrations of whole-brain 3D images, using the insight toolkit. The Insight Journal July–December
Yushkevich PA, Piven J, Hazlett HC et al (2006) User-guided 3D active contour segmentation of anatomical structures: significantly improved efficiency and reliability. Neuroimage 31:1116–1128
Fedorov A, Beichel R, Kalpathy-Cramer J et al (2012) 3D Slicer as an image computing platform for the Quantitative Imaging Network. Magn Reson Imaging 30:1323–1341
Kamnitsas K, Ledig C, Newcombe VFJ et al (2017) Efficient multi-scale 3D CNN with fully connected CRF for accurate brain lesion segmentation. Med Image Anal 36:61–78
Al-Rfou R, Alain G, Almahairi A et al (2016) Theano: A Python framework for fast computation of mathematical expressions, arXiv:1605.02688
Barber PA, Demchuk AM, Zhang J, Buchan AM (2000) Validity and reliability of a quantitative computed tomography score in predicting outcome of hyperacute stroke before thrombolytic therapy ASPECTS Study Group. Alberta Stroke Programme Early CT Score. Lancet 355:1670–1674
Chen L, Bentley P, Rueckert D (2017) Fully automatic acute ischemic lesion segmentation in DWI using convolutional neural networks. Neuroimage Clin 15:633–643
Maier O, Menze BH, von der Gablentz J et al (2017) ISLES 2015—a public evaluation benchmark for ischemic stroke lesion segmentation from multispectral MRI. Med Image Anal 35:250–269
Maldjian JA, Chalela J, Kasner SE, Liebeskind D, Detre JA (2001) Automated CT segmentation and analysis for acute middle cerebral artery stroke. AJNR Am J Neuroradiol 22:1050–1055
Nowinski WL, Gupta V, Qian G et al (2013) Automatic detection, localization, and volume estimation of ischemic infarcts in noncontrast computed tomographic scans: method and preliminary results. Invest Radiol 48:661–670
Gillebert CR, Humphreys GW, Mantini D (2014) Automated delineation of stroke lesions using brain CT images. Neuroimage Clin 4:540–548
Peter R, Korfiatis P, Blezek D et al (2017) A quantitative symmetry-based analysis of hyperacute ischemic stroke lesions in noncontrast computed tomography. Med Phys 44:192–199
Hampton-Till J, Harrison M, Kühn AL et al (2015) Automated quantification of stroke damage on brain computed tomography scans: e-ASPECTS. EMJ Neurol 3:69–74
Schramm P, Schellinger PD, Fiebach JB et al (2002) Comparison of CT and CT angiography source images with diffusion-weighted imaging in patients with acute stroke within 6 hours after onset. Stroke 33:2426–2432
Schramm P, Schellinger PD, Klotz E et al (2004) Comparison of perfusion computed tomography and computed tomography angiography source images with perfusion-weighted imaging and diffusion-weighted imaging in patients with acute stroke of less than 6 hours' duration. Stroke 35:1652–1658
Availability of data and materials
The trained deep learning models are available on reasonable request sent to the corresponding author.
This study received research and development funding from Helsinki University Hospital Medical Imaging Center.
Ethics approval and consent to participate
Ethical committee approved this retrospective study, and patients’ informed consent was waived.
Consent for publication
The authors have no conflicts of interest to declare.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Öman, O., Mäkelä, T., Salli, E. et al. 3D convolutional neural networks applied to CT angiography in the detection of acute ischemic stroke. Eur Radiol Exp 3, 8 (2019). https://doi.org/10.1186/s41747-019-0085-6
- Computed tomography angiography
- Convolutional neuralnetwork
- Machine learning
- Neural networks (computer)