- Technical note
- Open Access
Can portable tomosynthesis improve the diagnostic value of bedside chest X-ray in the intensive care unit? A proof of concept study
© The Author(s) 2017
- Received: 29 May 2017
- Accepted: 21 August 2017
- Published: 27 October 2017
Portable bedside chest X-ray (CXR) is an important and frequently used tool in the intensive care unit (ICU). Unfortunately, the diagnostic value of portable CXR is often low due to technical limitations and suboptimal patient positioning. Additionally, abnormalities in the chest may be hidden on the projection image by overlapping anatomy and devices such as endotracheal tubes, lines and catheters. Digital tomosynthesis (DTS) can solve the problem of anatomical overlap. In DTS, several low-dose X-ray images from different angles are acquired and subsequently used by a reconstruction algorithm to compute section images along planes parallel to the detector. However, a portable device to be used for portable bedside chest DTS is not on the market yet. In this work, we discuss modifications to a portable X-ray device to enable portable DTS and illustrate the potential of portable DTS to improve the diagnostic value of bedside CXR in the ICU. A simulation, based on computed tomography scans, is presented. Our experiments comparing portable DTS with conventional bedside CXR showed a substantially improved detection of pneumothorax and other abnormalities.
- Bedside chest X-ray
- Digital tomosynthesis
- Intensive care unit (ICU)
- Portable (mobile) technology
The diagnostic value of portable bedside chest X-ray (CXR) is often limited, especially in the ICU
Digital tomosynthesis (DTS) offers separation of anatomical overlap in portable CXR into different section images
A novel concept for a portable DTS device is described
Portable DTS exams of ICU patients were simulated, based on computed tomography scans
Simulated portable DTS showed potential to improve the diagnostic value of portable CXR
Chest X-ray (CXR) plays an important role in patient management in the intensive care unit (ICU) , assisting physicians in the diagnosis and follow-up of a variety of cardiopulmonary disorders. It is valuable for identification of findings necessitating emergency medical care, such as pneumothorax, (ventilator-associated) pneumonia or atelectasis, and assessment of volume status . Additionally, CXR is used to evaluate a broad range of intrathoracic medical devices, ensuring proper positioning and surveying for complications [3–5]. Although the clinical usefulness for daily follow-up of patients with CXR in the ICU is under constant scrutiny [6, 7], the American College of Radiology considers the use of CXR in the ICU highly appropriate, especially upon admission and after placing invasive devices such as endotracheal tubes, endovascular catheters and drains, or when the condition of the patient worsens .
Bedside CXR is the imaging modality of choice for evaluation of the chest in ICU-patients, since these patients need to be monitored closely and are often less mobile due to mechanical ventilation or other medical devices such as drains and tubes. However, the diagnostic value of portable CXR in the ICU is significantly lower when compared to a CXR obtained with a dedicated wall-mounted flat panel device. First of all, compared to a standard posterior-anterior (PA) and lateral (LAT) exam, the portable anterior-posterior (AP) exam of a bedridden patient in supine position shows an apparently enlarged width of the heart, which may obscure retrocardiac structures, especially in poor inspiration. Also, the detection and assessment of certain abnormalities (e.g. pneumothorax) is known to be more difficult on portable CXR obtained in a reclining patient when compared to CXR acquired from an upright patient . Additionally, the clinical condition of the patient often hinders optimal patient positioning, causing asymmetric images and superposition of anatomical structures (e.g. of the clavicles over the lung apices). Frequently, superposition of devices, tubes and catheters as well as patient related factors (obesity, hypoventilation, motion) further reduce the quality of the portable CXR in the ICU. Finally, the portable CXR is subject to technical limitations: images are acquired using a voltage of 90 kVp instead of 120 kVp, resulting in a less translucent appearance of the ribs. Image contrast in portable CXR is also reduced due to the absence of an anti-scatter grid while the reduced source-image distance of only 1.2 m causes higher geometrical distortions .
A portable device for DTS does not exist yet. In our simulation, we assume a portable X-ray modality is used to acquire several X-ray projections from different angles. The X-ray tube moves in a motorised way along a straight path. Alternatively, a system could be designed without motion, using a linear array of carbon nanotube sources. In our simulation, the source-image distance was set to 120 cm and the total tube travel distance was set to 14 cm. This corresponds to an angular range of 6° (substantially less than the typical 35° angular range in chest DTS with a dedicated wall-mounted flat panel detector) to reduce the risk of collision with bedside equipment and simplify the design of the device.
We retrospectively selected two ICU patients from the hospital Picture Archiving and Communication System (University Hospital of Antwerp) who showed abnormal findings on CXR and underwent a chest CT scan within a time frame of 1 h. CT scans had been obtained using a 64-slice unit (LightSpeed VCT, GE Medical Systems, US) with the following technical parameters: 99 mA, 120kVp, 600 ms. The need for informed consent was waived by the Ethical Committee. From the anonymised CT data, which consisted of voxels of 0.5 × 0.5 × 0.84 mm size, 15 DTS projections were simulated using the ASTRA tomography toolbox (v1.6, Antwerp, Belgium) , which contains algorithms for simulating X-ray radiographs from volumetric data. Simulated DTS projections were computed by a linear forward projection of the CT volume onto a virtual flat panel detector, and contained 831 × 1012 pixels of 0.42 mm size. No scatter or electronic noise were simulated. Next, from the simulated DTS projections, 55 coronal section images of 5-mm depth were reconstructed using 50 iterations with the simultaneous iterative reconstruction technique (SIRT) . Finally, the reconstructed DTS section images were compared to the original CXR and interpreted for ICU-related abnormalities. The effective dose of the portable chest DTS exam, as described in this study, was computed using the PCXMC  simulation software.
To validate this DTS simulation based on CT scans, a simulated DTS exam was compared with an experimentally acquired DTS exam of an anthropomorphic chest phantom (Humanoid Systems, Carson, USA). For the experimental DTS exam, 15 chest DTS projections were acquired manually (i.e. without motorised tube motion) with a mobile X-ray unit (Agfa Healthcare DXD-100, Mortsel, Belgium), from the same angles and source image distance as in the ICU simulations above. All DTS projections were obtained at 90 kVp and 0.1 mAs, which was the lowest possible tube current of the mobile unit. Subsequently, 55 coronal section images of 5-mm depth were computed by DTS using 30 SIRT iterations. For the simulated DTS exam, 15 DTS projections were simulated from a CT scan of the anthropomorphic phantom and a simulated DTS reconstruction was computed. Reconstructions from the experimental DTS exam and CT based simulated DTS exam were compared visually by a radiologist with 10 years’ experience in thoracic imaging.
In this work we describe a new modality for bedside chest DTS. In the presented simulations of ICU patients, portable DTS improved the detection of abnormalities in the bedside CXR. Whereas the clinical use of chest DTS for patients in the radiology department can be evaluated against CT, MRI and other modalities, far fewer alternatives exist for bedside chest examination of patients in the ICU, for whom transport to a radiology department is often complicated and expensive. The cost of performing bedside ultrasound as a daily follow-up is relatively high, since this exam takes longer to perform and requires a high level of expertise. Mobile CT scanners exist but are not widespread and they are mainly limited to dedicated applications such as imaging of the head [23–25]. Portable chest DTS could be implemented on a mobile X-ray device, which is already widely accepted as a mobile examination device. Portable DTS might also hold potential to improve bedside exams in the emergency room, sterile rooms in haematology wards or transplantation units, burn units, surgical departments, etc. In future work, a more elaborate prospective clinical study of potential applications will be conducted, in combination with an optimisation of the acquisition technique.
The radiation dose for a chest DTS exam using a dedicated wall-mounted flat panel is around 0.12 mSv for a typical acquisition of 60 projections . In our experiment, however, only 15 DTS projections were taken at 90 kVp and 0.1 mAs. The effective dose of 0.029 mSv corresponds to approximately 1.5 times the dose of a portable CXR, obtained at 90 kV and 1 mAs.
Söderman et al.  showed that decreasing the tube travel distance, and thus the angular range, has a positive effect on the reproduction of the trachea and paratracheal tissue, vessels, and aorta. However, a smaller angular range resulted in decreased image quality related to following vessels through the volume. Image quality for findings more specifically related to critically ill patients, such as pneumothorax/pneumomediastinum, and the capability of separating overlapping devices into different slices were not discussed. Also, the aforementioned study investigated different configuration parameters for chest DTS with an effective dose comparable to a standard DTS exam. Our simulations showed portable chest DTS in the ICU with a small angular scanning range (only 6°) and a radiation dose of only 0.029 mSv. The effective slice thickness of the simulations presented in this study was obtained experimentally by adding a thin line throughout the simulated patient from head-left-posterior to foot-right-anterior. The amount of this line that was visible in the reconstructed tomosynthesis slices illustrated an effective slice thickness corresponding to a patient slab of 30 mm. Despite this relatively high effective slice thickness, the diagnostic quality of the bedside CXR was improved.
The presented simulation study has obvious limitations. First, the quality of the bedside DTS reconstruction relies heavily on the accuracy of the measured relative positions and orientation of the X-ray source trajectory and detector, which was assumed to be error-free in our simulation. In a real-life acquisition, our previous work could be applied to correct possible misalignment [28, 29]. Secondly, patient motion during the acquisition of the DTS projections is another well-known cause of degradation of DTS image quality, which was not included in our simulation. Despite a reduced total acquisition time of a few seconds, due to the reduced number of 15 exposures, breathing or other types of patient motion might still occur. Many ICU patients might not be able to hold their breath, although for intubated patients, a short interruption in mechanical ventilation could be considered in some cases. Reconstruction and motion correction methods might need to be incorporated and improved in a portable chest DTS device [30, 31]. Finally, physical phenomena such as scatter and quantum noise were excluded from these preliminary simulations. However, the reconstructions from the experimentally acquired DTS exam of the humanoid phantom strengthen our belief that the conducted experiments are at least indicative of the expected performance of mobile chest DTS.
In conclusion, we have shown with preliminary simulations that portable chest DTS holds potential to improve the diagnostic accuracy of bedside CXR in the ICU. Possible benefits include: improved localisation of parenchymal consolidations (anterior versus posterior), detection of pneumothorax or pneumomediastinum patients in the supine position, verification of the correct position of drains and lines, differentiation between pleural effusions and consolidations, and other applications. It is technically feasible to perform mobile chest DTS with a modified mobile X-ray unit, which is already widely accepted as a mobile examination tool.
This study was supported by Agfa HealthCare and the Agency for Innovation by Science and Technology in Flanders (IWT).
JC drafted the paper and performed the simulations. AS provided the relevant clinical images for the simulation and performed the image interpretation. GB, PMP and JS critically revised the paper draft. All authors read and approved the final manuscript.
Ethics approval and consent to participate
The need for informed consent was waived by the Ethical Committee.
JC and GB are employees of Agfa HealthCare.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Bentz MR, Primack SL (2015) Intensive care unit imaging. Clin Chest Med 36:219–234View ArticlePubMedGoogle Scholar
- Rubinowitz AN, Siegel MD, Tocino I (2007) Thoracic imaging in the ICU. Crit Care Clin 23:539–573View ArticlePubMedGoogle Scholar
- Porté F, Basit R, Howlett D (2009) Imaging in the intensive care unit. Surg 27:496–499Google Scholar
- Godoy MCB, Leitman BS, De Groot PM, Vlahos I, Naidich DP (2012) Chest radiography in the ICU: Part 1, evaluation of airway, enteric, and pleural tubes. Am J Roentgenol 198:563–571View ArticleGoogle Scholar
- Godoy MCB, Leitman BS, De Groot PM, Vlahos I, Naidich DP (2012) Chest radiography in the ICU: Part 2, evaluation of cardiovascular lines and other devices. Am J Roentgenol 198:572–581View ArticleGoogle Scholar
- Cruz J, Ferra M, Kasarabada A, Gasperino J, Zigmund B (2014) Evaluation of the clinical utility of routine daily chest radiography in intensive care unit patients with tracheostomy tubes: a retrospective review. J Intensive Care Med 31:333–337View ArticlePubMedGoogle Scholar
- Graat ME, Stoker J, Vroom MB, Schultz MJ (2005) Can we abandon daily routine chest radiography in intensive care patients? J Intensive Care Med 20:238–246View ArticlePubMedGoogle Scholar
- American College of Radiology (2014) ACR appropriateness criteria: routine chest radiographs in ICU patients. Available via https://acsearch.acr.org/docs/69452/Narrative/
- Kollef MH (1991) Risk factors for the misdiagnosis of pneumothorax in the intensive care unit. Crit Care Med 19:906–910View ArticlePubMedGoogle Scholar
- Hobbs DL (2007) Chest radiography for radiologic technologists. Radiol Technol 78:494–516, quiz 517–519PubMedGoogle Scholar
- Dobbins JT, McAdams HP, Godfrey DJ, Li CM (2008) Digital tomosynthesis of the chest. J Thorac Imaging 23:86–92View ArticlePubMedGoogle Scholar
- Chou S, Kicska G, Pipavath S, Reddy G (2014) Digital tomosynthesis of the chest: current and emerging applications. Radiographics 34:359–372View ArticlePubMedGoogle Scholar
- Johnsson AA, Fagman E, Vikgren J et al (2012) Pulmonary nodule size evaluation with chest tomosynthesis. Radiology 265:273–282View ArticlePubMedGoogle Scholar
- Chawla A, Boyce S, Washington L, McAdams H, Samei E (2009) Design and development of a new multi-projection X-ray system for chest imaging. IEEE Trans Nucl Sci 56:36–45View ArticleGoogle Scholar
- Bertolaccini L, Viti A, Tavella C et al (2015) Lung cancer detection with digital chest tomosynthesis: 1rst round results from the SOS observational study. Ann Transl Med 3:67PubMedPubMed CentralGoogle Scholar
- Yamada Y, Jinzaki M, Hashimoto M et al (2013) Tomosynthesis for the early detection of pulmonary emphysema: diagnostic performance compared with chest radiography, using multidetector computed tomography as reference. Eur Radiol 23:2118–2126View ArticlePubMedGoogle Scholar
- Vult von Steyern K, Björkman-Burtscher IM, Höglund P et al (2012) Description and validation of a scoring system for tomosynthesis in pulmonary cystic fibrosis. Eur Radiol 22:2718–2728View ArticlePubMedGoogle Scholar
- Johnsson ÅA, Vikgren J, Båth M (2014) A retrospective study of chest tomosynthesis as a tool for optimizing the use of computed tomography resources and reducing patient radiation exposure. Acad Radiol 21:1427–1433View ArticlePubMedGoogle Scholar
- Molk N, Seeram E (2015) Digital tomosynthesis of the chest: a literature review. Radiography 21:197–202View ArticleGoogle Scholar
- Van Aarle W, Palenstijn WJ, Cant J et al (2016) Fast and flexible X-ray tomography using the ASTRA toolbox. Opt Express 24:25129–25147View ArticlePubMedGoogle Scholar
- Gregor J, Benson T (2008) Computational analysis and improvement of SIRT. IEEE Trans Med Imaging 27:918–924View ArticlePubMedGoogle Scholar
- Servomaa A, Tapiovaara M (1998) Organ dose calculation in medical X-ray examinations by the program PCXMC. Radiat Prot Dosim 80:213–219View ArticleGoogle Scholar
- McCunn M, Mirvis S, Reynolds N, Cottingham C (2000) Physician utilization of a portable computed tomography scanner in the intensive care unit. Crit Care Med 28:3808–3813View ArticlePubMedGoogle Scholar
- Carlson AP, Yonas H (2012) Portable head computed tomography scanner-technology and applications: experience with 3421 scans. J Neuroimaging 22:408–415View ArticlePubMedGoogle Scholar
- Masaryk T, Kolonick R, Painter T, Weinreb DB (2008) The economic and clinical benefits of portable head/neck CT imaging in the intensive care unit. Radiol Manage 30:50–54PubMedGoogle Scholar
- Bath M, Svalkvist A, von Wrangel A et al (2010) Effective dose to patients from chest examinations with tomosynthesis. Radiat Prot Dosim 139:153–158View ArticleGoogle Scholar
- Söderman C, Asplund S, Allansdotter Johnsson Å et al (2015) Image quality dependency on system configuration and tube voltage in chest tomosynthesis—a visual grading study using an anthropomorphic chest phantom. Med Phys 42:1200–1212View ArticlePubMedGoogle Scholar
- Cant J, Behiels G, Sijbers J (2016) Automatic geometric calibration of chest tomosynthesis using data consistency conditions. 4th International Conference on Image Formation in X-Ray Computed Tomography, Bamberg, 161–164. http://ctmeeting.shpci.org/data/ProceedingsCTMeeting2016.pdf
- Cant J (2017) Iterative reconstruction for mobile chest tomosynthesis. Dissertation, University of Antwerp. http://hdl.handle.net/10067/1406630151162165141
- Van Eyndhoven G, Sijbers J, Batenburg K J (2012) Combined motion estimation and reconstruction in tomography. 12th European Conference on Computer Vision, vol. 7583, Firenze, Lecture Notes on Computer Science, 12–21 https://link.springer.com/chapter/10.1007/978-3-642-33863-2_2
- Van Nieuwenhove V, De Beenhouwer J, De Schryver T, Van Hoorebeke L, Sijbers J (2015) Affine deformation correction in cone beam computed tomography. Fully three-dimensional image reconstruction in radiology and nuclear medicine, Newport, 182–185 http://www.fully3d.org/Previous_upcoming_meetings/website/download/3D15Proceedings.pdf